For cleaning disinfection of medical instruments made of metal, plastic, glass. Aldehyde-free With active agent Glucoprotamin®. Effective against bacteria (incl. TB) and yeasts Limited virucidal according to RKI recommendation of 01/2004 (DVV 2008). RKI-listed (A).
For disinfection of medical instruments.
|Our performance||Your advantage|
|VAH and RKI listed, effective against bacteria (incl. TB) and yeasts, limited virucidal.||Protects the user from infection risks through contaminated instruments|
|Contains Glucoprotamine||Unifies the positive properties of the established active agents such as efficacy, biodegradability and user-friendliness|
|Powerful formulation components||Good material compatibility even against sensitive materials such as flexible endoscopes Approved by Olympus. Very good cleaning performance, even with heavily stained instrument types|
|Low volatility, low foam||Suitable for ultrasonic bath applications, pleasant fragrance|
For cleaning and disinfection of non-critical medical devices made of metal, glass etc. and for cleaning disinfection of semi-critical and critical medical devices before mechanical reprocessing. For purely manual reprocessing, for terminal disinfection, we recommend using Sekusept aktiv, Sekusept forte and Sekusept forte S.
Prepare the solution by diluting with water of at least drinking water quality. Place the instruments in the solution immediately after use, ensuring that all surfaces are completely wetted. Clean the instruments following the manufacturer’s instructions, including the mechanism (e.g. brushing, wiping). After the specified contact time, remove the objects from the liquid and rinse with water of at least drinking quality. according to RKI recommendation is to make the solution fresh every day.
Material changes in polycarbonate, polysulfide and acrylic glass with continuous use cannot be ruled out, but can be largely avoided by sufficient rinsing after every application. Not suitable for instruments made from silicone. If flexible endoscopes are chemical-thermally reprocessed mechanically, a thorough rinsing of the endoscope must be done after the disinfecting cleaning, in order to avoid reciprocal action with the products in the machine. For purely manual reprocessing, for terminal disinfection, we recommend using Sekusept aktiv, Sekusept forte and Sekusept forte S.
Service life of unloaded application solution (in covered instrument basin): 14 days
A loaded application solution is to be made fresh every working day according to the RKI recommendation.
Active ingredients contained in 100g:
Slight color variations in the concentrate do not mean any loss of quality.
|Use (after commencement)||Shelf life|
For a correct storage of environmental hygiene and instrument disinfectants, we request you to observe the following: The original containers (canisters, buckets or bottles) with the disinfectant concentrate (or powder) should be stored at temperatures below 25°C and kept dry and clean. A brief period below or above the given storage temperature (e.g. during transport) will not diminish the effectiveness.
Firmly seal the package after every use. Please also take care that the concentrate remaining in the container is not contaminated when preparing the application solution.
Assuming that the prepared application solutions are not affected by organic material and are stored in a clean, closed package or a covered instrument basin.
The containers must be appropriately marked in which the disinfectant solutions are stored, e.g. content, concentration, date, shelf life and batch number. Used and organically contaminated solutions are to be changed at least once per workday as per RKI recommendation.